Compact easy to use thermometer with large display. Auto field calibrate to ±2˚F in less than 10 seconds using ice water.
- Reversible head for best viewing
- Field calibrateable to ±2°F
- Easy-to-read LCD
- Data Hold for hard to reach areas
- Penetration tip
- Auto off
- 3-year limited warranty
USA Suggested List Price
To measure temperature with this unit follow these simple instructions:
- With the unit “OFF” press and continue to hold the ON/ OFF button.
- The LCD will alternate between °F and °C. Release the ON/OFF button when desired measurement mode appears on the LCD
- Remove the protective sheath from the stem.
- Press the ON/OFF button.
- Insert stem into area or object to be tested. At least 1/2” of the stem must be inserted.
Warning: The stem is a sharp and pointed object which can be dangerous.
The Data Hold function is used by pressing the D-H button when taking a measurement. The measurement will flash on the LCD when Data Hold is in use. To return to normal function, press and release the D-H button a
Press the ON/OFF button to end operation, or the unit will automatically power off after 35 minutes.
To reverse the LCD to suit your viewing angle:
Insert a small coin into notch in curved area “B”, then gently pry portion “A” from “B”. Shielded wires will keep the two parts connected. Twist the area “A” 180° and snap back into area “B”.
Get more accurate readings by using the right tip for your thermometer:
- Penetration: Use for immersion and air; air response time will be slower than if using anactual air probe.
- Chisel: Need chisel tip for surface temperatures. Penetration and/or air tips will not givean accurate surface reading.
- Air: Samples air temperatures three times faster than penetration or contact tips.
- Contact: Measures three times faster than chisel tip.
|Model 310C Specifications|
|Temperature Range||-58°F to 300°F (-50°C to 150°C)|
|Sample Rate||1 reading every 1.5 seconds|
|°C / °F Selectable||Yes|
TPI Pocket Thermometers Demonstration Video
Model 310C Applications
- Ambient air
- Calibrate thermostats
- Compressor heads
- Grill and surface temperatures
- Serving temperatures
Frequently Asked Questions
How do I check calibration of my pocket thermometer?
Place thermometer in a solution of crushed ice and water, swirl the water around and temperature should read close to 32°F.
Does the whole stem need to be immersed to get an accurate reading?
The sensor is in the tip of the probe and needs to be 1/2 inch into whatever you are measuring.
What battery does my pocket thermometer use?
The LR44 button battery is used in TPI pocket digital thermometers.
Click below to view or download product documentation PDFs: